In this video I am calibrating the Gophert CPS-3205C power supply using my Keithley 175 4.5 digit multimeter and my adjustable dummy load.
If you haven't watched the review click here:
The calibration procedure is:
1. Set the V/A selector switch to A position,
2. Push the adjust knob then power on the power supply,the meter will display PUSH OFF,
3. Press the ON/OFF button one time then press the LOCK button two times, the unit will go into voltage calibration (please note calibrate the voltage with no load), the voltmeter displays 01.00 and amp meter displays 001.
4. please adjust the adjust knob to set the output voltage to 1.005V,( you can push the adjust knob to select the range of adjusting) then press the ON/OFF button to calibrate the output voltage to 10.00V and 30.00V, after 30V voltage has been calibrated, please press the ON/OFF button to go into current calibration, connect your electronic load to the output terminal of the power supply, the unit must operate in C.C mode,the amp meter displays 1.00 and the voltmeter displays 0001, please adjust the adjust knob to set the output constant current to 1.000A, then press the ON/OFF button to calibrate the output constant current 5.000A,after 5.0A has been calibrated then press the ON/OFF button to finish the calibration.